MTM Scientific, Inc: How to Measure the Power Output of Solar Panels
A photovoltaic solar electric panel generates DC power when it is exposed to sunlight. A natural question to ask is 'How much power?' Here we explain how to test a solar panel for output and how to determine the maximum power point.
A solar panel is a generator of DC electrical power. DC electrical power wattage is the product of voltage and current.
DC Watts = Voltage X Current
Power is measured in units of Watts, (Symbol W)
Voltage is measured in units of Volts, (Symbol V)
Current is measured in units of Amps, (Symbol I)
The power generated by any solar panel in full sunlight depends on the resistance of the electrical load connected to it. Resistance is measured in units of Ohms. The relationship between current, voltage and resistance is referred to as Ohm's law. The current through a resistor is the applied voltage divided by the resistance.
Current = Voltage / Resistance
The method of measuring the power output of a solar panel is to connect resistors of various values to the panel and measure the voltage. The measurements can be used to calculate the power output. The same measurements can be used to plot the power output and create a performance graph for the panel.
A digital multimeter for measuring DC voltage is required for this test. An assortment of power resistors is also needed. In practice, each power resistor is connected to the panel and the voltage is measured with the meter, as shown in the drawing.
The meter is also used to measure the open circuit voltage and the short circuit current. Here is a table of data from a set of measurement on 3 large solar panels connected in series. In this example power resistors of 100, 50, 25 and 3 ohms were used. The current through each resistor is calculated by dividing the measured voltage by the resistance. The power is calculated by multiplying the voltage times the current.
Note that no power is generated when the open circuit voltage and short circuit current is measured. A more intuitive view of the data can be gained by plotting a graph of the voltage versus current.
The output of this solar panel shows a characteristic behavior that is common to all solar panels. The maximum power is generated at the operating point which forms the 'knee' in the curve. In our case, this is at approximately 81 watts, corresponding to a load resistance of 25 ohms. The maximum power point is where the product of current and voltage is a maximum. Expressed graphically, the maximum power point it where the largest area rectangle can be formed beneath the curve.
Testing a solar panel requires an assortment of power resistors capable of handling the power. In most cases, ordinary 1/4 and 1/2 watt resistors are NOT suitable for this type of measurement. Small resistors overheat and burn out. Power resistors are made to handle more wattage, although they will become hot over time and can burn out too. Adjustable power resistors (called rheostats) are also useful, although they require a meter for measuring the resistance after each adjustment. The simplest way to avoid problems with overheating is to use high wattage resistors and to make individual measurements as quickly as possible.
The power output of solar panels can be increased by tracking the position of the sun, as described in our Solar Tracker E-Booklet.
Electric DC gearhead motors can be used as a DC generator, and can be tested in a similar way, as described in our Motor Generator E-Booklet.
Special Offer: Power Resistor Assortment
MTM Scientific offers an assortment of POWER RESISTORS (CAT# POWER) for testing small solar electric panels. Our assortment contains 10 different fixed power resistors, each with a 5 Watt rating. The assortment also includes an adjustable 200 ohm power rheostat rated for 50 Watts. The fixed resistance values included are: 820, 100, 50, 27, 10, 5, 2, 1, .5 and .1 ohms. Each power resistor is individually labeled with the resistance value to make testing quick, easy and accurate. The cost of this item is $23.50 and it is sent postage paid in the USA. Please click the link below to purchase using Paypal. International Customers: Please scroll down for a special Paypal link which includes International Postage.
BY PAYPAL: click the "PAYPAL, Buy Now" button. You can change quantity at Paypal checkout to order more than one. Mail Orders: Please see details below.
Sorry: This item has Sold Out... Check Back Soon!
MAIL ORDER & INTERNATIONAL ORDER INFORMATION
We keep ordering simple! Prompt, courteous 30 day refund policy.
BY MAIL....to order by mail, please send a Check or Money Order for $23.50 to:
MTM Scientific, Inc. P.O. Box 522, Clinton, Michigan 49236 U.S.A.
Your item will be shipped by 1st Class Mail in the U.S.A.
International Customers: Your business is welcome, however we must charge an additional $12.00 ($35.50 total) to help defray the cost of Air Mail postage. (This item is heavy.) Please use the Paypal link below for International Orders. We also accept money orders in $USD sent by mail to the address above.
PAYPAL LINK: Power Resistor Assortment with International Postage included: Catalog Item #POWER-INTSorry: This item has Sold Out... Check Back Soon!
Click Here to View other Products from MTM Scientific, Inc.